View and Download YOKOGAWA EJXA installation manual online. EJX Series and EJA-E Series Differential Pressure and Pressure Transmitters. EJXA. Manual. Yokogawa Electric Corporation. EJX Series. HART Communication Type .. This manual describes the HART protocol communica- 1 EJXA Get Yokogawa EJXA Manual. Get all Yokogawa manuals!.
|Published (Last):||23 March 2006|
|PDF File Size:||1.45 Mb|
|ePub File Size:||10.27 Mb|
|Price:||Free* [*Free Regsitration Required]|
Minimum voltage limited at I nstall the transmitter at most h mm above the HP process connection Mounting the Flushing Modify structure of Table 1.
The following sorts of symptoms indicate that transmitter may not be operating properly. ERR with burnout direction switch. Please do not ground the shield on the RTD side of the cable. Select the type of valve most appropriate for the application.
YOKOGAWA EJX110A Installation Manual
Normally, only non- conductive pollution occurs. Use a slotted screwdriver to turn the zeroadjustment screw. Page 47 Large output error. Please read it along with the standard manuals. Page 81 Unit: Add HART 7 manual. Page 21 2.
YOKOGAWA EJXA INSTALLATION MANUAL Pdf Download.
I nstall the transmitter at least h mm below the Figure 3. Page 54 9. Page 3 ii 5. Grounding is always required for the proper operation of transmitters. Before mounting the transmitter to the process flange, affix the teflon film as follows: Page 4 iii 8. Model Page 46 8. II Pressure Transmitters Category: To prevent such problems, the process pressure taps must be angled as shown in figure 5.
Maintenance of the transmitter is easy due to its modular construction.
Absolute pressure and gauge pressure transmitters 49 pages. Page 85 ii Edition 7th 8th Date Feb. Got it, continue to print.
Page 11 b Never apply a voltage exceeding V DC V DC with an internal lightning protector for the insulation resistance test, nor a voltage exceeding V AC V AC with an internal lightning protector for the dielectric strength test. Add to my manuals Add. Page 55 Revision Information: Use a slotted screwdriver to turn the zero-adjustment screw. The zero point adjustment can be The local push button on the integral indicator must made with a resolution of 0.
Yokogawa EJXA : Manual
Note that powering off within thirty seconds after setting will cause a return to the previous settings. Zero adjustment Conduit connection 4. Add g and h. Flange mounted differential pressure transmitters 54 pages.
Connect a communicator and check self-diagnostics.